To have a better experience, please upgrade your IE browser.upgrade
Questo sito utilizza cookie di profilazione (propri e di terze parti) per ottimizzare la tua esperienza online e per inviarti pubblicità in linea con le tue preferenze. Continuando a utilizzare questo sito senza modificare le tue preferenze acconsenti all’uso dei cookie. Se vuoi saperne di più o negare il consenso a tutti o ad alcuni cookie clicca qui>
The website that you are visiting also provides Arabian language. Do you wish to switch language version?
يوفر موقع الويب الذي تزوره المحتوى باللغة العربية أيضًا. هل ترغب في تبديل إصدار اللغة؟
The website that you are visiting also provides Russia language Do you wish to switch language version?
Данный сайт есть в английской версии. Желаете ли Вы перейти на английскую версию?
The OptiX OSN 6800 is configured with the WMU. Wavelength lockout is performed for the OTU. After a period of time when the OTU is configured on the WMU, the WMU reports the WAVE_LEN_OVER alarm. OTU monitoring is deleted from the WMU, and hard reset is performed on the OTU. On the second day, the drift of the wavelength of the OTU is still serious.NE version: V100R
2. Query the tube core temperature of the two OTUs, that is, query the central frequency by running :optp:1,1,83,1,ff,3d,1,1,0;. The information "#9-1660:szhw [1660-Renhe][2009-09-19 10:48:00+08:00]>" is returned.
Optp cmd : ff3d
01 01 00 00 00
The last two bits of the tube core temperature of the OTU are in direct proportion to the wavelength.
3. Set the tube core temperature manually. Set the central frequency of the OTU to an offset. Not that the offset should be a small value. The WMU can lock only the offset within ±10 GHz.
4. Start wavelength search once manually after querying the wavelength through the MCA.
5. After one minute, use the MCA to query the central wavelength. The query result shows that the offset is larger. It indicates that the offset becomes larger after the adjustment.
6. Run :log-query:1,"bb10.log"; to query the black box of the board. (Note that you must initialize the black box before running the command. Otherwise, there are too many records to be viewed.)
7. Check whether the adjustment is valid based on the records in the black box. Start adjustment for several times manually. View the black box of the OTU. If there is no record in the black box, it indicates that the adjustment occurs on other OTUs. If there are multiple adjustment records about the OTU, but the wavelength drift in the WMU performance is still large, it is possible that the wavelength detected by the WMU is not transmitted by the OTU.
8. The preceding operations determine that the WMU can be adjusted. The offset, however, becomes larger after the adjustment.
9. Check the physical fiber connection. The check result shows that the physical fiber connection is proper. Compare the physical fiber connection diagram with the logical fiber connection. The result shows that the logical fiber connection is improper. Re-configure the logical fiber connection, and the WMU lockout is normal.
When the WMU is not configured, the WMU can adjust the central wavelength of the OTU. Why?The wavelength is adjusted according to the logical fiber connection. When the WMU is not configured, the adjustment is performed according to the logical fiber connection. When the WMU is configured with wavelength monitoring, but the configuration is inconsistent with the logical fiber connection, the OTU is adjusted incorrectly. Pay attention to this point during maintenance.
1. The OTU fault causes the drift of the wavelength.
2. The WMU fails. As a result, wavelength information cannot be detected correctly, or wavelength adjustment information cannot be reported to the SCC correctly.
3. The fiber connection error causes incorrect adjustment.