No relevant resource is found in the selected language.

# Why is the optical power measured by EXFO OSA different from that measured by huawei OSA on OPU-MON

## Issue Description

The engineer measure the optical power on OPU-MON by 3 different devices:
1、Huawei OSA (-8.37dBm).the attached file "Huawei OSA.bmp"
2、Optical power meter (-8.4dBm)
3、EXFO OSA (-12.69dBm). the attached file "EXFO OSA.bmp"
The signle optical power measured by the Huawei OSA and Power Meter is almost the same.
But the total optical power measured by EXFO OSA is different from the total optical power measured by huawei OSA on OPU-MON. Why?

Null

## Handling Process

Huawei OSA calculate the total optical power of signal and noise.
So the total optical power equal the optical power measured by Optical power meter.
EXFO OSA only calculate the total optical power of signal.We can calculate the total optical power of signal according Pawg on the diagram.
-21.14+10lg7＝－12.69dBm
When OSNR is very high (Noise is very low) and there are many wavelengths, the total optical power measured by huawei OSA and EXFO OSA  will be almost the same.

## Root Cause

The total optical power measured by huawei OSA and EXFO OSA have different consideration for noise.

## Suggestions

The total optical power need to cosider noise affection.

END